How to Set the Pitch Tone Through the SpO2 Sensor On Any CardioTech Patient Monitor
- On the main display, click into the ECG section to bring up the ECG setup menu
- Change “HR from” to SpO2.
- Go to the SpO2 section to bring up the SpO2 setup menu to make sure the pitch tone is turned on. You can also increase or decrease the volume of the pitch tone in this menu.